|Place of Origin:||China|
|Minimum Order Quantity:||1 Unit|
|Packaging Details:||Carton package|
|Product Name:||DOF6000-P Portable Open Channel Ultrasonic Flow Meter||Output:||Pulse,4-20mA), RS485/Modbus,Datalogger, GPRS|
|Degree Of Protection:||IP68||Weight:||3 KG|
|Data Logger:||16GB||Power Supply:||50 Hours Batterty|
open channel flow measurement devices,
open channel water flow meter
The DOF6000 series flowmeter consists of Flow calculator and the Ultraflow QSD 6537 Sensor.
The flow calculator can calculate the cross-sectional area of partially filled pipe, open channel stream or river, for stream or river, with up to 20 coordinate points describing the river’s shape of cross section. It is suitable for various applications
|Power supply:||Calculator:85-265VAC (Charging battery)|
|IP class:||Calculator: IP66|
|Operating temperature:||0℃ ~+60℃|
|Case material:||Fiber Glass|
|Display:||4.5” color LCD|
|Output:||Pulse,4-20mA(Flow & Depth),RS485/Modbus, Datalogger, GPRS|
|Application:||Partially Filled Pipe: 150-6000mm; Channel: width >200mm
|Velocity:||Velocity range:||20mm/sec to 12 m/sec. Bidirectional velocity capability.|
|Velocity accuracy:||±1% measured velocity|
|Range:||20mm to 5000mm (5m)|
|Accuracy:||± 1% of measured|
|Range:||0mm to 10000mm (10m)|
|Accuracy:||± 1% of measured|
|Temperature:||Range:||0°C to 60°C|
|Electrical Conductivity (EC):||Range:||0 to 200,000 µS/cm, Typically ± 1% of measurement|
|recorded as a 16-bit value (0 to 65,535 µS/cm) or a 32-bit value (0 to 262,143 µS/cm)|
|Range:||±70° in roll and pitch axes.|
|Accuracy:||±1° for angles less than 45°|
|Output:||SDI-12:||SDI-12 v1.3, Max. cable 50m|
|RS485:||Modbus RTU, Max. cable 500m|
|Environmental:||Operating temperature:||0℃ ~+60℃ water temperature|
|Storage temperature:||-20℃ ~+60℃|
|Others:||Cable:||The standard cable is 15m, the maximum option is 500m.|
|Sensor material:||Epoxy-sealed body, Marine Grade 316 Stainless Steel Mounting Bracket|
|Sensor size:||135mm x 50mm x 20mm (L x W x H)|
|Sensor weight:||1kg with 15m of cable|
The Ultraflow QSD requires very little maintenance. During routine site visits the following checks can be performed:
Piezo element faces
Clean the instrument surfaces where the Piezo elements are located by wiping with a cloth. If needed a plastic scraper can be used to remove any bio-fouling. Take care not to scrape the surface of the instrument. Refer to the above diagram for the areas where the ultrasonic eyes and conductivity sensors reside. These areas must be kept clean.
Pressure depth sensor
Check that the opening to the depth pressure sensor is clear of any fouling. Use a brush to clear any material.
With a cloth wipe the electrode faces. Do not use abrasive materials to clean them as this will affect the calibration of the conductivity measurement.
Inspect the cable to ensure that it has not been damaged.
Visually check the instrument to see that it has not been damaged by heavy debris in the measured stream.
With years of export experience along with excellent quality, advanced services and competitive prices, Lanry has won numerous customers' trust and support.With years of export experience along with excellent quality, advanced services and competitive prices, Lanry has won numerous customers' trust and support.
High Accuracy Low Cost Wall Mount Flow Meter standard meter: 3-7 days
Special meter: pls contact us
Express, by air, by sea can be optional.
Contact Person: Ms. Tracy Xu